Lookup NU author(s): David Langton,
Professor Tom Joyce
Full text for this publication is not currently held within this repository. Alternative links are provided below where available.
There have been no large comparative studies of the blood levels of metal ions after implantation of commercially available hip resurfacing devices which have taken into account the effects of femoral size and inclination and anteversion of the acetabular component. We present the results in 90 patients with unilateral articular surface replacement (ASR) hip resurfacings ( mean time to blood sampling 26 months) and 70 patients with unilateral Birmingham Hip Resurfacing (BHR) implants (mean time 47 months). The whole blood and serum chromium (Cr) and cobalt (Co) concentrations were inversely related to the size of the femoral component in both groups (p < 0.05). Cr and Co were more strongly influenced by the position of the acetabular component in the case of the ASR, with an increase in metal ions observed at inclinations >45 degrees and anteversion angles of <10 degrees and >20 degrees. These levels were only increased in the BHR group when the acetabular component was implanted with an inclination >55 degrees. A significant relationship was identified between the anteversion of the BHR acetabular component and the levels of Cr and Co (p < 0.05 for Co), with an increase observed at anteversion angles <10 degrees and >20 degrees. The median whole blood and serum Cr concentrations of the male ASR patients were significantly lower than those of the BHR men (p < 0.001). This indicates that reduced diametral clearance may equate to a reduction in metal ion concentrations in larger joints with satisfactory orientation of the acetabular component.
Author(s): Langton DJ, Sprowson AP, Joyce TJ, Reed M, Carluke I, Partington P, Nargol AVF
Publication type: Article
Publication status: Published
Journal: Journal of Bone and Joint Surgery: British Volume
ISSN (print): 0301-620X
Publisher: British Editorial Society of Bone and Joint Surgery
Altmetrics provided by Altmetric