Lookup NU author(s): Dr Anders Barlow,
Dr Jose Portoles,
Dr Naoko Sano,
Professor Peter Cumpson
This is the authors' accepted manuscript of an article that has been published in its final definitive form by Cambridge University Press, 2016.
For re-use rights please refer to the publisher's terms and conditions.
The development of the helium ion microscope (HIM) enables the imaging of both hard, inorganic materials and soft, organic or biological materials. Advantages include outstanding topographical contrast, superior resolution down to <0.5 nm at high magnification, high depth of field, and no need for conductive coatings. The instrument relies on helium atom adsorption and ionization at a cryogenically cooled tip that is atomically sharp. Under ideal conditions this arrangement provides a beam of ions that is stable for days to weeks, with beam currents in the order of picoamperes. Over time, however, this stability is lost as gaseous contamination builds up in the source region, leading to adsorbed atoms of species other than helium, which ultimately results in beam current fluctuations. This manifests itself as horizontal stripe artifacts in HIM images. We investigate post-processing methods to remove these artifacts from HIM images, such as median filtering, Gaussian blurring, fast Fourier transforms, and principal component analysis. We arrive at a simple method for completely removing beam current fluctuation effects from HIM images while maintaining the full integrity of the information within the image.
Author(s): Barlow AJ, Portoles JF, Sano N, Cumpson PJ
Publication type: Article
Publication status: Published
Journal: Microscopy and Microanalysis
Print publication date: 01/10/2016
Online publication date: 13/09/2016
Acceptance date: 15/08/2016
Date deposited: 24/08/2016
ISSN (print): 1431-9276
ISSN (electronic): 1435-8115
Publisher: Cambridge University Press
Altmetrics provided by Altmetric