Lookup NU author(s): Professor Thomas Scharf
This work is licensed under a Creative Commons Attribution 4.0 International License (CC BY 4.0).
Background: Design processes such as human-centered design, which involve the end user throughout the product development and testing process, can be crucial in ensuring that the product meets the needs and capabilities of the user, particularly in terms of safety and user experience. The structured and iterative nature of human-centered design can often present a challenge when design teams are faced with the necessary, rapid, product development life cycles associated with the competitive connected health industry.Objective: We wanted to derive a structured methodology that followed the principles of human-centered design that would allow designers and developers to ensure that the needs of the user are taken into account throughout the design process, while maintaining a rapid pace of development. In this paper, we present the methodology and its rationale before outlining how it was applied to assess and enhance the usability, human factors, and user experience of a connected health system known as the Wireless Insole for Independent and Safe Elderly Living (WIISEL) system, a system designed to continuously assess fall risk by measuring gait and balance parameters associated with fall risk.Methods: We derived a three-phase methodology. In Phase 1 we emphasized the construction of a use case document. This document can be used to detail the context of use of the system by utilizing storyboarding, paper prototypes, and mock-ups in conjunction with user interviews to gather insightful user feedback on different proposed concepts. In Phase 2 we emphasized the use of expert usability inspections such as heuristic evaluations and cognitive walkthroughs with small multidisciplinary groups to review the prototypes born out of the Phase 1 feedback. Finally, in Phase 3 we emphasized classical user testing with target end users, using various metrics to measure the user experience and improve the final prototypes.Results: We report a successful implementation of the methodology for the design and development of a system for detecting and predicting falls in older adults. We describe in detail what testing and evaluation activities we carried out to effectively test the system and overcome usability and human factors problems.Conclusions: We feel this methodology can be applied to a wide variety of connected health devices and systems. We consider this a methodology that can be scaled to different-sized projects accordingly.
Author(s): Harte R, Glynn L, Rodríguez-Molinero A, Baker PMA, Scharf T, Quinlan LR, Ó'Laighin G
Publication type: Article
Publication status: Published
Journal: JMIR Human Factors
Print publication date: 16/03/2017
Online publication date: 16/03/2017
Acceptance date: 12/01/2017
Date deposited: 05/06/2017
ISSN (electronic): 2292-9495
Publisher: JMIR Publications, Inc.
PubMed id: 28302594
Altmetrics provided by Altmetric